How to use Pocket Diagnostic® plant disease tests
The following page describes the simple process of using Pocket Diagnostic® lateral flow plant disease tests for the in-field testing of Phytophthora, Erwinia amylovora, Potato virus Y and Ralstonia solanacearum.
- Select sample (as detailed below)
- Cut or tear sample into small pieces and put into bottle containing buffer
- Shake firmly for 30-60 seconds to break up the sample and allow the liquid to settle
- Draw liquid into the pipette – avoid sample debris and air bubbles
- Keeping the test device level, add 2 drops into the sample well of the device
- Valid results within 10 minutes
Click to enlarge
Always read the instructions provided with the kits before use.
Select symptomatic plant material from where the healthy material meets the diseased material. As a general rule, we recommend that the sample should weigh approx. 0.2g or 25x25mm in size. It’s important to wash any soil and other debris from the plant material sample prior to adding to the bottle.
Selection of the most appropriate sample ensures accurate and reliable results.
The unique ‘bottle and ball’ sample preparation method sufficiently breaks down plant material after only 30-60 seconds of shaking. After shaking the sample should be left to settle prior to a few drops being removed from the top of buffer and added to the tests device.
This easy and clean approach enables Pocket Diagnostic® plant disease tests to be used on the following plant material:
- Flowers (cut into small pieces)
- Bark (cut into small pieces)
- Woody material (cut into small pieces)
Hard materials may require slightly longer shaking to ensure adequate sample breakdown.
One line present only at the C line indicates a negative result.
Two lines present, at both the T and C line indicates a positive result.
The absence of a line at C indicates the test has failed and must be repeated.
Click to enlarge
Example of a test used incorrectly
Below is an example of what can happen if too much debris and plant material are added to the test strip. There is too much debris preventing the test from working properly.
Flooded sample well and viewing window
Always read the instruction before using Pocket Diagnostic®.